Labview Error Device Identifier Is Invalid
Refer to user documentation for more information. Error -200211Multiple relay names were specified for a single relay operation. You must open the Configuration panel and associate the NI PXIe-5603 with a supported or external digitizer and a supported or external LO. Error -200244Hardware clocking error occurred. Check This Out
Bring in your external sample clock through one of the higher-frequency front panel connectors, or use a lower sample rate. Attachments: Report Date: 12/22/2004 Last Updated: 11/20/2015 Document ID: 3GL8JLBP Your Feedback! The acquisition has stopped, therefore the sample specified by the combination of position and offset will never be available. Disable multiplexed excitation, or use one of the physical channels between ai0 and ai7.
Change the range or the data. Refer to documentation for information about the maximum length allowed for user-defined information. Verify that any accessories configured with this device are connected. If you are using an external sample clock or an external reference clock, make sure it is connected and within the jitter and voltage specifications.
Do I need to include that file into the build? Error -200070The NI PXI-5665 (14 GHz) is not configured properly. Error -200078Analog input (AI) task started or committed during a counter 0 DMA acquisition. If you are generating your clock internally, please contact National Instruments Technical Support.
Error -200069Device specified is not a valid switch device. Only the leading portion was saved. Error -200076Current data supplied is outside of the specified range. https://forums.ni.com/t5/Multifunction-DAQ/DAQmx-Device-identifier-is-invalid/td-p/1857925 Error -200206Advance trigger and Advance Complete event must use the same polarity in this particular switch device.
Thanks once again. For example, instead of using a voltage task for DevX on analog input channels 1 and 2, you need to use DevX/ai1:2. Error -200193Switch device does not support this operation. Choose a frequency within this range.
Set the buffer size to 0, and/or change the data transfer mechanism to Programmed I/O. Error -200267Product of AO channel properties Maximum Value and Gain exceeds the maximum voltage for the device. I was able to solve the problem. Cannot configure this port or any of its lines for handshaking at this time.
Once it’s opened up, you should see your myDAQ listed as NI myDAQ “name”. http://jvmwriter.org/labview-error/labview-error-ni-488.html Reduce your sample rate. Error -200237Hardware clocking error occurred. Error -200223Specified threshold and hysteresis values for this channel create a triggering range that is not supported by your device.
Error -200159Requested operation could not be performed because the digital lines are being used for communication with SCXI or a TEDS carrier. Everything in the build process went smoothly however when I try to run the application I get the "200220" error stating my device identifier is invalid however I made sure of Regards, Share this post Link to post Share on other sites rccardude909 0 LAVA groupie Members 0 3 posts Posted October 10, 2008 Hey Minh Yes I installed the latest http://jvmwriter.org/labview-error/labview-error.html Error -200045EEPROM of the device appears to be corrupt.
DAQmx Error: Device identifier is invalid. The following constraint must hold: Offset > (Gain * Minimum Value / 2) Error -200269Specified Offset is too large for the given AO Gain and Maximum Value. Error -200281Reading relative to the reference trigger or relative to the start of pretrigger samples position before the acquisition is complete.
If you need to use the same physical channel more than once inside your list of channels, refer to that physical channel under different names.
Select another device with an available internal excitation source or supply external excitation. Error -200038"Wait" instruction cannot be the last instruction of a "repeat until" loop. You can also use a product with more onboard memory or reduce the number of programs your computer is executing concurrently. Error -200075Voltage data supplied is outside of the specified range.
You must open the Configuration panel and associate the NI PXIe-5603 with a supported or external digitizer and a supported or external LO. Error -200036Number of iterations specified for a finite "repeat" loop is invalid. For optimal performance, use a buffer size that is a multiple of 8 times the sector size. http://jvmwriter.org/labview-error/labview-error-7.html The rate was coerced to the slowest possible convert rate.
Use Measurement & Automation Explorer. Error -200271Product of AO Channel properties Minimum Value and Gain exceeds the minimum voltage for the device. Cannot configure this port or any of its lines for input at this time. Error -200063The NI PXIe-5601 is not associated with a digitizer or an LO.
If possible, use counter 0 instead of counter 1. Lower the gain, or adjust AI Minimum and/or AI Maximum. Device performance is not guaranteed. Error -200099Physical channel not specified.
Error -200003Channel cannot be used more than once inside a list of channels. Thanks. Change the direction of the task, use another terminal, or use another device. Error -200040Source terminal to be routed could not be found on the device.
Error -200082Minimum is greater than or equal to the maximum. Sign In Sign In Remember me Not recommended on shared computers Sign in anonymously Sign In Forgot your password? Error -200177Operation is permitted only while the switch device is scanning. To unroll a loop, remove the "repeat" and "end repeat" instructions and explicitly replicate the instructions of the removed loop the desired number of times.