Labview Error Code 50405
I suspect there is a problem with the external electronics i'm using. Contact us about this article Hi All, I am pretty newbie to Labview. Issue 1 Restoration Options Close the open driver session (niScope_close or DAQmxClearTask) and open a new one to the device (niScope_init or DAQmxCreateTask). This is a simple description for the above code. 1) DAQ Assistant can make a DO signal. http://jvmwriter.org/labview-error/labview-error-code-9.html
If that's successful, try importing that task into LabVIEW or running an example program. Bei mir lag der Fehler auf der Hardwareseite. Showing results for Search instead for Did you mean: Reply Topic Options Subscribe to RSS Feed Mark Topic as New Mark Topic as Read Float this Topic to the Top Bookmark Does this error occur in Measurement & Automation Explorer (MAX) Test Panels? http://digital.ni.com/public.nsf/allkb/AE7F2281D910D0E286257950005C5131
Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase EnglishSpanishJapaneseKorean 16 ratings: 2.12 out of 5   DAQmx Error -50808 Primary Software: Driver Why is this happening? The light of the board is off and to make it work again I have to plug it in and out. Mit der Nutzung des Forums erklärst Du dich damit einverstanden, dass wir Cookies verwenden.Es wird in jedem Fall ein Cookie gesetzt um diesen Hinweis nicht mehr zu erhalten.
I may not be perfect, but I'm all I got! 0 Kudos Message 3 of 9 (2,942 Views) Reply 0 Kudos Re: Error -50405 on USB 6008 Windows 7 Brian_P Member It's easy! Please tell us why. The only thing that chnges is the time that the card works, but after some time, usually less than one hour it crashes.
KnowledgeBase 3NLF8T7G: USB Host Controller Chipsets for use with M Series USB / NI CompactDAQ KnowledgeBase 4D4BO9AY: Why is My USB Data Acquisition Device Not Responding? When the machine starts, the hot solonoid valves are opened, cold closed, and the zone valves rotated to direct water to the hot sump tank. From: Brian F.
shall I use another hardware to control of input data? I dont know how you going to add functionalities nad features in the future. Ist das ein intelligenter Weg? This is because when i execute the program with the electronics unplugged, the loop executes very well without any problems.
Most of the time, the program works fine. Is that external USB hub powered? KnowldegeBase 4VBA46G0: C Series Digital I/O Behavior on Loss of Ethernet, Wireless, or USB Connection Attachments: Report Date: 11/22/2011 Last Updated: 09/20/2016 Document ID: 5RLB14FS Your Feedback! Using a decent cable?
Number 2 and 3 is the method which is manually processed by user. his comment is here Sign in here. Swap the USB cable if you can. Here is the problem.
I am writing data from each sensor in individual text file simultaneously. Not in a noisy RF environment? Register a new account Sign in Already have an account? this contact form Index(es): Date Thread Flag as inappropriate (AWS) Security UNIX Linux Coding Usenet ArchiveAboutPrivacyImprint newsgroups.derkeiler.com >Archive >Comp >comp.lang.labview >2007-04 HOME | SEARCH | REGISTER RSS | MY ACCOUNT | EMBED RSS |
Any help is appreciated. KnowledgeBase 664ADVJ0: Why Is My USB DAQ Device Not Recognized in Device Manager When Connected through a USB 3.0 Port? Connecting the device to a different host controller or replacing your USB cable may help decrease the likelihood of transaction errors.
Windows Users can configure whether the USB device can be used or not.
Applications EngineerNational Instruments 2 Kudos Message 4 of 9 (2,932 Views) Reply 2 Kudos Re: Error -50405 on USB 6008 Windows 7 computali Member 08-05-2011 01:43 AM Options Mark as New Open Measurement& Automation Explorer (MAX) and expandDevices and Interfacesto locate your device. Any driver calls to the current session(s) may result in the following symptom: Error -1074130544 occurred: The session handle is not valid. error 50405 Mole.Luca Member 04-04-2012 08:17 AM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator Now I am working
tas | tas murah | tas cantik 0 Kudos Message 7 of 9 (2,709 Views) Reply 0 Kudos Re: Error -50405 on USB 6008 Windows 7 Jake_H Member 08-08-2011 12:15 PM The chip uses SPI protocol to send data and the chip acts as SPI master. Answered Your Question? 1 2 3 4 5 Document needs work? navigate here error 50405 NI-DAQ 6008 stops acquiring.
Error -50808 : It occurred when you used DAQmx 9.4 or above. Gruß Julius Empfehlungen: expressionflow, LavaG , mooregoodideas, OpenG, JKI Blog Tipp 13.08.2009, 08:51 Beitrag #5 gottfried LVF-Guru Beiträge: 1.631 Registriert seit: Mar 2007 2012 2004 EN 20** Oesterreich Error -50405 Ich After 90 seconds, the hot solonoids close, cold open, and the zone valves switch position. Using a decent cable?
All rights reserved. | United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Most Active Hardware Boards : Multifunction DAQ : Error -50405 on USB 6008 Windows Share this post Link to post Share on other sites CArnold 0 LAVA groupie Members 0 3 posts Posted October 9, 2008 Thanks, I tried the help/explain error and am Error -50405 :It occurred when you used DAQmx 9.3.5 or below. I am doing a LabView project that needs to acquire data at 30MHz from a chip in debugging mode.
Good luck JR Share this post Link to post Share on other sites fuzzycontrolfreak 0 Active Members 0 18 posts Version:LabVIEW 2009 Since:2009 Posted July 8, 2007 thankyou very much Right on one of the root hubs on the computer, or does it go through an external USB hub? Showing results for Search instead for Did you mean: Reply Topic Options Subscribe to RSS Feed Mark Topic as New Mark Topic as Read Float this Topic to the Top Bookmark Es werden 10 Messungen gemacht und danach 10sek gewartet.
Thankyou so much for your help though ... Reset the device from the open driver session (niScope_ResetDevice or DAQmxResetDevice) Open Measurement& Automation Explorer (MAX) and expand Devices and Interfaces to locate your device. Ich kommuniziere auch mit einem Gerät (über TCP/IP), welches ab und zu mal (relativ selten) nichts zurücksendet. You can also try to programmatically reset the USB DAQ device.
Thankyou so much for your help. One cycle consists of 90 seconds of 190F water followed by 90 seconds of 70F water. Sign In Sign In Remember me Not recommended on shared computers Sign in anonymously Sign In Forgot your password? Disconnect and reconnect your device from the system.
If anyone could tell me what possible reasons could result in this error, so i can track the problem directly, because it seems very complicated to know exactly where the problem The machine has four hot solonoid valves, four cold solonoid valves, two zone valves for directing water back to either a hot sump or cold sump, a hot water heater maintained