Labview Error 88705
I owe you a drink!” Cheri: - 7 Months Ago “I was getting loads of errors until I tried this. Abhinav T.Applications EngineeringNational Instruments IndiaLabVIEW Introduction Course - Six HoursGetting Started with NI-DAQmxMeasurement Fundamentals 0 Kudos Message 2 of 4 (4,800 Views) Reply 0 Kudos Re: Error -88705 Device not found, No error in VI mode. >>> >>> LabVIEW Real Time 7.1 >>> PXI-8186 Controller >>> PXI-6281 M series A/D card >>> SCXI-1102B module, using PXI-6281 as digitizer. >>> >>> Physical channels This issue should be addressed in an upcoming release of NI-DAQmx. http://jvmwriter.org/labview-error/labview-error-ni-488.html
Repair Labview Error 88705 Posted: This is a suprisingly common error, and I have a Repair! Please tell us why. The problem seems to be related to the nidevldu since it refuses to be stopped or restarted. Then rename Data folder from C:\Program Files\National Instruments\MAX\ directory. internet
A workaround for this problem is to add some code, after the first invocation of the DAQmx VI that returns the -88705 error, to recognize this error, invoke the Wait VI Right-click on theNI Device Loaderservice (also seen as nidevldu), and selectPropertiesfrom the drop-down menu. NOTE: If the download link doesn't work you may need to Download it Directly from a Mirror Here. STEP 3: Click the "Repair All" Button to Repair Your PC!
By downloading and running the registry repair tool RegCure Pro, you can quickly and effectively fix this problem and prevent others from occuring. It has done this 1 time(s). Changing this configuration should ensure that the correct services start every time you reboot your computer. You could wrap the DAQmx VI and the Wait VI in a for loop to accomplish this.
My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. See KnowledgeBase 4BHHID6DE: Why Are NI Device Loader (nidevldu) and NI Configuration Manager Services Not Starting? Don't Worry - I'm here to help you fix it! http://forums.ni.com/t5/PXI/Error-88705-Device-not-found-not-present-PXI-DaqMX/td-p/523465 Worked a treat!” Cherelle- 1 Month Ago “Thanks for sharing, I no longer have to put up with the dreaded Labview Error 88705” Sommer- 2 Months Ago “Will this work with
Yes, after installing the DAQ software the first time i tested it and the selftest passed. To update the firmware, run the utility FWUpdate from any desktop PC, located here:
KnowledgeBase 491FPHB9: USB Device Firmware Incompatibility with DAQmx Base 3.0 or Later KnowledgeBase 5XG8M1RP: Why Does My DAQ or Simulated DAQ Device Fail the MAX Self-Test With Error -88705? get redirected here I am waiting for your answer.Have a nice day. 0 Kudos Message 8 of 34 (2,572 Views) Reply 0 Kudos Re: Error 88705 with a USB-6366 DAQ BenHabegger Member 02-07-2012 07:47 This can also occur randomly as an internal software error. Showing results for Search instead for Did you mean: Reply Topic Options Subscribe to RSS Feed Mark Topic as New Mark Topic as Read Float this Topic to the Top Bookmark
It took a bit longer than 10 minutes, closer to 30, but by the time it was finished my PC worked great again. http://jvmwriter.org/labview-error/labview-error.html I booted to save mode, disabled all NI services and then uninstalled NI software which suceeded. The Windows Windows Labview Error 88705 are easy to repair. To start it properly go to Start >> Control Panel >> Administrative Tools >> Services and select the NI Device Loader Service.
I tried this before already. That's wonderful media since it implies that labview error 88705 odds are excellent that labview error 88705 your difficulty continues to be well documented and can probably be sorted out on Click here to get the free tool. http://jvmwriter.org/labview-error/labview-error-7.html How can I resolve this issue?
For example, DAQmx Base 3.1 for Windows is only compatible with NI-DAQmx 8.6 for Windows. Why do I receive this error and how do I make Measurement & Automation Explorer recognize my device? The following corrective action will be taken in 0 milliseconds: No action. " The fix posted by Abhinav doesn't seem to work, as the device no longer even appears on
Start MAX and attempt a Self-Test.
I hope this helps. I also unsinstalled all NI - Software and reinstalled it, no luck. Most errors on your machine are caused by uninstalling programs, installing new ones and accidentally deleting important files. My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
Below are descriptions and potential solutions to some common errors seen when using a USB device with the NI-DAQmx Base driver. In MAX, NI-DAQmx Devices, performing the Self Test yields an error message "Error -88705 occurred at Self Test. ...The specified device is not present or not active in the system." However, At that point, the workaround will no longer be necessary. his comment is here Since first install, twice my device stopped being recognised (also faild selt test with -88705), with errpr event in Windows event log: " "The NI Device Loader service terminated unexpectedly.
A >> fatal status of -88705 indicates that the driver for the specified >> device is not present. >> > Yes. Related Links: KnowledgeBase 42HG08DD: Why Do I Get a Measurement & Automation Explorer (MAX) Database Connection Error? Causes of the error: Windows Windows Labview Error 88705 are caused by misconfigured system files. The service has the state "started" .
Is it a problem in the new MAX version?? Luckily, the vast majority of these prospective concerns are unusual. NI-DAQmx 7.3.0 is installed on that PXI. Primary Software: Driver Software>>NI-DAQ Primary Software Version: 4.6.1 Primary Software Fixed Version: N/A Secondary Software: Driver Software>>NI-DAQmx Problem: I have a DAQ device or simulated DAQ device in Measurement & Automation
To Fix the problem you need to follow the 3 steps : STEP 1: Download & Install RegCure Pro for Free. Keila Says: at 4:48 AM worked just like it said.