Labview Error 50202
Dec 2007: Re: Import/.Net Controls to Palette... In an internet search for this tag, I found one link to nipalsm.exe. Jan 2008: M-I'5`Persec ution - wh y won' t the Briti sh police do their j ob an d put a s top to it? I won't have a chance to test it with a real setup the next weeks... http://jvmwriter.org/labview-error/labview-error.html
URL: http://labview360.com/forum/forum_posts.asp?TC=04OBOCR9KK82 使用DELTA IFD6500(USB to RS485) 與DELTA Temperature controller DTA4848V1連線(更新使用說明) (PLC主機 - 台達電) ... 的很好~有問題可以提出一起研究!! PCIe slots?0 0 02/04/14--01:11: Does NI DAX assistance can be used using NI cRIO 9075 Contact us about this article Hello... I am using Labview 2012 My hardware consists of a 9174 Chassis, 9401 IO and 9205 AI. I still don't know what in your code could possibly cause this error, but being able to pinpoint it to a function or subVI with some probes or error indicators would try this
URL: http://labview360.com/forum/forum_posts.asp?TC=GM4009IVGISJ Labview waveform chart的X軸時間會一直跑掉 (LabVIEW) ... Best regards, Peter 0 Kudos Message 2 of 9 (1,544 Views) Reply 0 Kudos Re: DAQmx - error 50202 crossrulz Knight of NI 02-11-2014 02:48 PM Options Mark Before this, I used ALLOW REGENERATION property to conitnuously generate waveform that was a nice feature 'cause i don't need to write every time the new waveform. So now we know we are looking at NI Platform services.
Calling DAQmx Start for both of your tasks outside the loop would reduce memory fragmentation and significantly improve performance. It points to Time clock issues. I was wondering if it's even possible to take measurements in RSE or NRSE mode? It's currently installed in a PXIe 1082 chassis with a PXIe 8101 controller. Logically they should be indicators, and since comparisons with floating point numbers can cause problems, I recommend fixing the code to convert all the wires to blue and eliminate the coercion
Please clarify your suggestion. Any troubleshooting advice would be great! I see your Write Data to Text File uses timestamps. Then you use a PC with the development environment and use the Connect to a Remot Panel under the Operate menu to get to it.
You may also want to look at the memory consumption during the run (doesn't hurt).JLV 02. That tag is called pal. ERROR by somch... 31. Dec 2007: Re: Dynamic creation of controls on the fly by avni ... 31.
Jan 2008: M'I`5,Perse cution ' Berna rd L evin expre sses h is vie ws by mimfv... 01. Dec 2007: ??? Plot 3dsurface Line and X,Y,Z Axis .... They are best explained in the following KnowledgeBase articles.How Do I Handle Error 50202 in LabVIEW or MAX?
what is the considerations for using third party DAC (such as loading or .... ) Thanx0 0 02/04/14--20:24: different sampling rate Contact us about this article Hi, his comment is here Did you go into that one and see anything relevelant? Low Cost Keypad Test.vi 115 KB State Machine TypeDef.ctl 5 KB 0 Kudos Message 9 of 32 (566 Views) Reply 0 Kudos Re: What is causing error 50202 RavensFan Knight of comp.lang.labview [Top] [AllLists]
JLV ______________________________________________________________________ 0 Kudos Message 2 of 7 (1,433 Views) Reply 0 Kudos Re: Error 50202 MB1 Member 10-23-2007 07:57 AM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed The flow of the program is completely time-based, and I'm using an "Elapsed Time" VI to control the timing for each step.I didn't think upgrading to 8.5 would help, but I've Dec 2007: Re: wiener filter VI by Gaura... 31. http://jvmwriter.org/labview-error/labview-error-7.html Jan 2008: Re: IAI driver by TOO x... 01.
United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Most Active Software Boards : LabVIEW : What is causing error 50202 LabVIEW Register for the community · URL: http://labview360.com/forum/forum_posts.asp?TC=Z0AGF8O90FS8 多波型在Graph上，X軸如何自訂 (LabVIEW作業習題研討區) ... 請問各位大大：我將2個陣列顯示在同一個Waveform Graph上，現在我想改變我X軸的初始值 OR X軸的值也讓我自己輸入?2個件事分開做得到，但2個動作我結合不起來，請高人指點，感謝各位。X軸測試.vi感謝各位大大關鍵在於Waveform Graph傳入內容的特性Waveform Graph是LabVIEW中較特別的 ... It can't be terribly old, considering you're using LV 2010, but there may have been memory leaks fixed in the last year.
United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Most Active Software Boards : LabVIEW : Error 50202 LabVIEW Register for the community · Log in ·
I am just pretesting the configuration. Thank you in advance.0 0 02/03/14--09:10: ni cdaq 9184 earth ground Contact us about this article Hello, i wonder what is, exactly, I did an explain error on -50202 on my PC and got this message. "Error -50202 occurred at an unidentified location Possible reason(s): NI Platform Services: A system call returned Go to Solution. 0 Kudos Message 1 of 9 (1,627 Views) Reply 0 Kudos Re: DAQmx - error 50202 NI_PSi Member 02-11-2014 02:30 PM Options Mark as New Bookmark Subscribe Subscribe Best regards, Peter 0 Kudos Message 6 of 9 (1,513 Views) Reply 0 Kudos Re: DAQmx - error 50202 flobsi_1 Member 02-12-2014 06:45 AM Options Mark as New Bookmark Subscribe Subscribe
The application sequentially fires outputs with NI 9485 modules and checks input voltage on a 9201 module, and then appends data to a text file. Related Links: ASUS BIOS 1010KnowledgeBase 4BFBEIQA: AMD Multi-Core/CPU Systems Can Cause Unexpected Behavior in NI Driver Software Attachments: Report Date: 08/14/2007 Last Updated: 05/11/2011 Document ID: 4CDAP7QA Your Feedback! I have attempted to make the changes you suggested and attached my latest code. http://jvmwriter.org/labview-error/labview-error-ni-488.html I hope you find this information useful.
Simply use DAQmx Read in a loop and fetch a number of samples resulting in about 100ms toacquire this number of samples (per channel). 96kS/s --> 960 Samples to read per October 2007 ✉ ❰ ❱ Have you looked at these threads?http://forums.ni.com/ni/board/message?board.id=250&message.id=13974 http://forums.ni.com/ni/board/message?board.id=280&message.id=1022 << unlikely (?) If you notice that it "crashes" after approximately the same amount of time, it would sound So just put in a wait for as long as you need (probably an event structure with a timeout so you can abort). The buffer that is being set is the buffer in the card.
Dec 2007: M.I, 5`Persecu tion . by ZhN n... 31. I am getting this error on a deployment machine; it does not have labview development software. The test is supposed to run for about 6 days total.
URL: http://labview360.com/forum/forum_posts.asp?TC=ZCOI6ROZZ85K 上一頁 下一頁 找不到您要的資料嗎? 列印格式 | 寄給朋友 | 加到最愛 | 意見回饋 | 加到知識庫 | 開啟知識庫 | | |我的最愛 |隱私權保護政策 |免責聲明 |廣告刊登 |系統整合 |合作提案 |聯絡我們 |電話： 0988-002692 您可以在以下的頻道跟LabVIEW360互動： Facebook: Dec 2007: M`I,5.Persecut ion - MI5 H ave Systematical ly Destr oyed My Life by ievem... 31.