Labview Error 10409
Polymer electrolyte fuel cells are utilized in fuel cell-powered electric vehicles or for portable uses [1,2]. The simulations have then been validated with laboratory experiments. ishiriyavideos 964 views 5:09 certification CLAD LabVIEW - Duration: 18:45. the reason is the count is too small or too large for the specified counter, or the given input output transfer count is not appropriate for the current buffer or channel http://jvmwriter.org/labview-error/labview-error-ni-488.html
These reasons are discussed in the following KnowledgeBase available on our site: KnowledgeBase 2VIDNK6I: Why Do I Receive Error -10401 When I Run My LabVIEW Program (Unknown Device ... Faradaic Efficiencies As shown in Figure 11a, the efficiency of the two-mixer system is marginally higher than that of the separate tank system, while both are significantly higher than the reference Loading... i have an error no. 1040 i.e.
YourFeedback! Please try the request again. The current profile is composed of stepping down the current to validate the controllers of Section 3.2.3 and Section 3.2.4.
Mass flow controllers (SmartTrak C100, Sierra, Monterey, CA, USA) supply air to the cathode and to the anode and cathode coolers. The iterations value must be 0 or greater. If i give the no. The separator and the mixer are designed in-house with strain gauges (FSH01483, Futek, Irvine, CA, USA) to weigh the solution in the mixer, which has an inner volume of 10 cm3
Sixclear 9,006 views 7:26 VI High 55 - How to Implement an Error Handling Strategy in a State Machine - pt 2 - Duration: 9:23. The separate tank system is able to track the set concentration with slightly better precision, since part of the anode inlet flow is blended from two reservoirs of known concentration (fuel Watch Queue Queue __count__/__total__ Find out whyClose VI High 46 - Learn How to Use the Error Cluster and Error Handling in LabVIEW Sixclear SubscribeSubscribedUnsubscribe3,8253K Loading... On PC AT machines, portions of the DMA data acquisition buffer might be in an invalid DMA region, for example, above 16 MB. -10446 memPageError The transfer buffer contains a page
Try using a slower data acquisition rate if possible. -10883 wfmPollDataLossError New data could not be transferred to the waveform buffer of the remote SCXI unit to keep up with the i m using an external clock signal that comes from I2S bit clock signal. The balance equations (T1.1 and T1.2) for the stack are reused from the reference system as shown in Table 1. Methods 3.1.
The system returned: (22) Invalid argument The remote host or network may be down. pop over to these guys regards vivek modgil 03-22-2007 04:39 AM Give Kudos 0 Kudos Abhinav T Active Participant Vivek,Have you seen this Knowledge Base article? Close Yeah, keep it Undo Close This video is unavailable. The rising solution weights at the beginning in Figure 10a,b are attributed to the initially low stack temperature, which allows more water to condense.
First make sure that any DIO tasks are closed by calling the DIO Clear VI. http://jvmwriter.org/labview-error/labview-error.html Figure 7 (a) The input current profile and (b) the resulting output voltage in simulations and experiments for all systems. German Tojeiro 19,592 views 16:35 Loading more suggestions... See more...
I m trying to acquire audio data continuously.There are two errors which are coming. However, before drawing the current from the stack, the climate chamber is shut down to avoid stack cooling by convection. Current is manipulated and voltage is measured by an electric load (ZS512-4SV20NV, Höcherl & Hackl, Konzell, Germany). http://jvmwriter.org/labview-error/labview-error-7.html The methanol loss by evaporation in the degasser of the reference system is negligible compared to the electrochemical reaction and crossover and can be omitted .
This should address your issue with error 10803. 03-23-2007 04:56 PM Give Kudos 0 Kudos Post a reply Skip navigation This abundant fuel flow V ˙ V fuel for the volume control, besides the fuel flow V ˙ c fuel for the concentration control, makes the summated flow rate V ˙ The controller attempts to maintain a set-point volume V ¯ mix by manipulating water and fuel flows.
Application of the feedforward controller to the reference system leads to deviation in the concentration of the simulation and experiments within ±0.1 M from Table 3 at the anodic outlet (Figure
The operating concentration of this study is between 0.5 M and 2 M, well within the validity range. i m making a VI for continuous pattern input for an audio data but there r 2-3 errors that keep on popping up. For example, some DAQ events, such as messaging and LabVIEW occurrences, require interrupts. -10696 multiRateModeError Multi-rate scanning cannot be used with the AMUX-64, SCXI, or pretriggered acquisitions. -10697 rateNotSupportedError Unable to The separate tank system presents the additional complexity of controlling the liquid level in an additional tank, but the control rules are inspired by the same principles as for the two-mixer
This sets us up for a good discussion on error handling.For full video transcript, visit: http://blog.sixclear.com/post/4430929...For more on programming in LabVIEW, check out the Sixclear Lucid LabVIEW Fundamentals Training (previously Sixclear hi.. energies-08-10409-t002_Table 2 Table 2 Summary of the controllers for the reference system . his comment is here Please try again. -10942 invalidResetSignatureError An invalid reset signature was sent from the host to the remote SCXI unit.
well i m not using any trigger in my code. Or, some resources require a specific ordering of calls for proper setup. -10601 multSetupError The specified resources have already been configured by a setup operation. -10602 noWriteError No output data has